Function of Cardiac Stabilizer--Fixed heart in the process of cardiac
surgery. The epicardium and surface tissue of the operative site have
been fixed to provide stable surgical fields for operation.
The new stable Cardiac stabilizers made by Beijing medos AT
Biotechnology Co., Ltd, which are designed for Off-pump CABG
with powerful fixation and simple use.
outstanding stability and passed CE Certify. They are exported to
USA, GERMANY, RUSSIA, KOREA and other countries,
while market share in China is No.1.
Stable-V1 Vacuum Stabilizer
Stable-MV1 Micro Vacuum Stabilizer
Stable-M1 Mechanical Stabilizer
Stable-P1 Heart Positioner
Five kinds of fixed heads can be chosen randomly
Two kinds of adaptive retractor
Perfect fixed head
360 degree omni-directional rotation
Fixed head and connect arm form an angle of 90 degree
Flexible arm assembly
Extension cardiac surgery operation space
Achieve the best stabilizing effect
Convenient self-locking function
Retractor suture platform
Multiple groups of surgical sutures can be suspended at the same time
Reduce the amount of suture and provide clear surgical field
For Medtronic & Chase For Medos & Maquet
|Vacuum stabilizer||Stable-PV1-1||BM4001(Maquet)||Normal Vacuum Stabilizer+Tubing set+Retractor Platform|
|Vacuum stabilizer||Stable-WV1-1||BM4004(Medtronic)||Universal Vacuum Stabilizer+Tubing set|
|Micro Vacuum stabilizer||Stable-PMV1||BM4008 (Maquet)||Normal Micro Vacuum Stabilizer+Tubing set+Retractor Platform|
|Micro Vacuum stabilizer||Stable-WMV1||BM4007(Medtronic)||Universal Micro Vacuum Stabilizer+Tubing set|
|Mechanical stabilizer||Stable-PM1||BM4003 (Maquet)||Normal Mechanical Stabilizer+Retractor Platform|
|Mechanical stabilizer||Stable-WM1||BM4002(Medtronic)||Universal Mechanical Stabilizer|
|Heart Positioner||Stable-PP1||BM4006 (Maquet)||Normal Heart positioner+Tubing set+Retractor Platform|
|Heart Positioner||Stable-WP1||BM4005 (Medtronic)||Universal Heart positioner+Tubing set|
Instruction for Use
1.Building in retractor.
2.Before assemble the stabilizer ensure the interlink arm is flexible.
3.Assemble the stabilizer onto the platform by pulling back on the mount lever and hooking the mount onto the platform in the desired location. Lock the mount into place by sliding the mount lever 90ºto the left or right.
4.Attach one end of the tubing to the stabilizer tubing. Attach the other end of the tubing to a fluid collection canister (not include).Make sure th e stopcock is in the closed air position.The remaining vacuum tubing should then connect from the fluid collection canister to a regulated vacuum sou rce.Set the vacuum regulator from 200mmHg to 400mmHg.
5.Gently shape malleable feet as desired to conform to the heart without bending the feet beyond 25 degree in any direction. Place the stabilizer foot onto the heart and apply vacuum by opening the stopcock.
6.Once positioned, secure the arm by turning the mount knob clockwise.
7.To remove stabilizer from the retractor,release the vacuum by turning the stopcock to air.Loosen the stabilizer by holding the arm with one hand and turning the knob counter-clockwise with the other hand.
8.To remove stabilizer from the platform,slide the mount lever to the center position and pull back.
9.Unhinge the mount to release the stabilizer from the platform.
10.To remove the platform,fully close the system by turning the drive handle clockwise.Gently remove the system from the sternum.
11.To remove the platform from an Activator drive Mechanism,turn release latches on the platform in the direction of the arrows and pull the platform away from the drive mechanism.